The 5-in-1 weather station is compatible with My AcuRite:
|Sensor Installation||5-in-1 sensor installation|
|Icon Definitions||Click here|
|Replacement Parts||Replacement Display: 06058M
Replacement Lightning Detector: 06045M
5-in-1 PRO Weather Sensor: 06014RM
5-in-1 Weather Sensor: VN1TXC
PRO+ Upgrade for the 5-in-1 Sensor: 5N1DSSA
Replacement 5-in-1 Sensor Wind Cups: 5N1WC / 06031RM
Replacement 5-in-1 Sensor Wind Cups Assembly (with Bearings): 5N1WCBSA
Replacement 5-in-1 Sensor Wind Vane: 5N1VANE
Replacement 5-in-1 Sensor Mounting Bracket: 5N1MB
Mounting Tripod for Environmental Sensors: 06055M
Dimensions: 5.5-inch Height x 7.6-inch Width x 0.6-inch Depth
LIGHTNING DETECTORDimensions: 5.875-inches Height x 2.625-inches Width x 0.9-inches Depth
Wireless Range: 330 feet / 100 meters depending on home construction materials
Lightning Detection Range: 1 to 25 miles (1.6 to 40 kilometers)
Temperature Range: -40º to 158º F, -40º C to 70º C
Humidity Range: 1% to 99% RH
Data Reporting: 24 second updates during normal conditions; 8 second updates once lightning detected
Power: 4 AA alkaline or lithium* batteries
*Lithium batteries recommended for temperatures below -4 degrees Fahrenheit / -20 degrees Celsius
5-IN-1 WEATHER SENSOR
Outdoor Temperature Range: -40º to 158º F, -40º C to 70º C
Temperature Accuracy: +/- 2 degrees Fahrenheit; Learn More
FAQ & Troubleshooting
|Outdoor temperature is flashing or showing dashes||
If you are not receiving a signal from the sensor, or the temperature is flashing, showing dashes, or stuck, the unit may be experiencing wireless interference.
Perform a hard reset:
1. Bring both the sensor and display unit indoors and position them side by side.
Read more about wireless interference and extending the wireless range here: http://www.acurite.com/wireless-signal-range
No outdoor sensor reception
If the wireless signal indicator shows no bars:
|Inaccurate temperature or humidity||
|Inaccurate wind readings
|Display screen not working
Items Needed: 5-in-1 sensor, Philips screwdriver, damp cloth, car wax, toothbrush (optional), moth ball (optional)
|| Rain Gauge Calibration Demo
The rain gauge on the 5-in-1 sensor can be calibrated to improve accuracy by following the directions below.
Items Needed:5-in-1 sensor, display console, plastic cup, pin, screw driver
If the rain gauge doesn't register close to 1.06" after the cup is empty of water, then you should make an equal adjustment to the two calibration screws on the bottom of the 5-in-1 sensor. Be sure to adjust both screws the same amount. Turning screws clockwise will increase rainfall amount. Turning screw counter clockwise will decrease it.
There should be nearly an equal number of water drops (typically about 25 water drops) between bucket tips. If not, then adjust the calibration screws on the bottom of the 5-in-1 sensor until an equal number of water drops are tipping the buckets. Then, restart the calibration procedure.
If you don't hear the buckets tipping and see water dripping alternately out of each drain, there may be an issue with the rain gauge or it's adjustment. Contact Customer Support.
|How to mount the 5-in-1 sensor
Questions? We're here to help.
Phone +1-262-729-4852 | My AcuRite Technical Support +1-262-729-4850 | Community forum: http://support.acurite.com
Models: , 01024M, 06058M, 01049M